AirSense™ 10 is one of the world’s most widely used series of CPAP and APAP machines, each designed to deliver high-quality therapy for a better night’s sleep.
Build your new sleep system.
Build your new sleep system.
With each AirSense 10 device offering built-in cellular technology, your clinical team has access to therapy data and remote access to your device settings.
Every AirSense 10 machine includes a wide range of features designed to deliver comfortable therapy, as well as myAir access to help you stay on track throughout your sleep apnea treatment journey.
Let’s take a look at the differences between CPAP, Elite and Autoset models.
AirSense 10 CPAP and Elite
These are fixed-pressure machines, meaning they deliver a consistent pressure that is set by your doctor
AirSense 10 AutoSet
Auto-adjusting machines change airflow pressure levels throughout the night based on your needs.
The AutoSet algorithm is one of the most clinically published in the field of sleep-disordered breathing. It works to automatically adjust to each patient’s therapy pressure as their needs change to deliver the patient’s ideal, lowest therapy pressure.
Your AirSense 10 machine has been set up for your needs by our team. But it’s understandable you may want to make small adjustments based on your comfort needs. Here are some tips on how to adjust the comfort settings.
How to access your comfort settings on the AirSense 10:
On your AirSense 10 machine, rotate the dial to highlight My Options and press the dial to see your current settings. From here, you can personalize your options.
The humidifier moistens the air and is designed to make your therapy more comfortable. With the Climate Control Auto setting, both temperature and humidity are automatically preset at levels designed for maximum comfort. If you have a ClimatelineAir ™ heated tube, all you have to do is plug it in and press Start. Note: ClimatelineAir heated tubing is required to use Climate Control Auto.
You can also manually set the Humidity Level to Off or between 1 and 8, where 1 is the lowest humidity setting and 8 is the highest humidity setting.
How to adjust humidity:
Ramp Time is the period during which the pressure increases from a low start airflow pressure to your prescribed treatment pressure. You can set your Ramp Time to Off, 5 to 45 minutes or Auto. When Ramp Time is set to Auto, the device will detect when you have fallen asleep and then gradually increase to the prescribed treatment pressure.
How to adjust Ramp Time:
Need more help with your comfort settings? Your medical equipment store can help make any other necessary adjustments.
You should clean your machine weekly.
How to clean your CPAP machine
Time it takes: 5 minutes
Supplies you’ll need to clean your machine:
Weekly step-by-step cleaning tips:
ResMed has observed that prolonged and repeated use of an ozone device may eventually cause internal damage to a ResMed machine, leading to an increase in motor noise. The ResMed limited warranty excludes damage to ResMed machines caused by use of ozone devices.
It’s important to regularly inspect, clean and replace your CPAP supplies which can be affected by dirt and general wear. You should regularly check the water tub, air tubing and the air filter for any wear or damage.
Here are some replacement tips for your CPAP parts:
While ResMed masks are designed to be compatible with most ResMed machines, it’s best to confirm compatibility for your specific mask to determine the correct mask setting for your machine.
Here’s a list of recommended compatible masks:
Full face masks:
Nasal/nasal pillow masks:
Search by product name, type or brand.